VI Agilent U1251A/U1252A User’s and Service Guide. CAUTION † Turn off the circuit power and discharge all high-voltage capacitors in the circuit before. you perform resistance, continuity, diodes, or capacitance tests. † Use the correct terminals, function, and range for your measurements.
Electronics service manual exchange : schematics,datasheets,diagrams,repairs,schema,service manuals,eeprom bins,pcb as well as service mode entry, make to model and chassis correspondence and more.
The U1252A can be used as a programmable square wave generator that allows you to stimulate electronic circuits for debugging purposes. Uncompromising ruggedness and safety Use The U1250A series handheld digital multimeter with confidence in almost any environment. NCP1252A Datasheet, NCP1252A PDF, NCP1252A Data sheet, NCP1252A manual, NCP1252A pdf, NCP1252A, datenblatt, Electronics NCP1252A, alldatasheet, free, datasheet ...
For U1251A, input impedance is 10 M Ω (nominal) for 5 V~1000 V ranges. For U1252A, input impedance is 10 MΩ (nominal) in parallel with 1.1 MΩ at dual display.  The accuracy could be 0.05%+10 for U1251A and 0.05%+5 for U1252A. Always use NULL function to zero out the thermal effect b efore measuring the signal. Agilent U1251A/U1252A Handheld Digital Multimeter Data Sheet 99 Washington Street Melrose, MA 02176 Phone 781-665-1400 Toll Free 1-800-517-8431 Visit us at www.TestEquipmentDepot.com